Martindale VI13800 Voltage Indicator
VI13800 Voltage Indicator
Security policy (edit with Customer reassurance module)
Delivery policy (edit with Customer reassurance module)
Return policy (edit with Customer reassurance module)
The standard relating to 2 pole voltage indicators is BS EN61243-3. All voltage indicators on the British & European market should comply. The latest version of the standard is BS EN61243-3:2010 and came fully in to force in May 2013.
The Martindale VI13700 has been widely used for many years and as part of its design has always included a fuse in the probe to limit the current in the event of damage to the cable.
There is a technical argument that fuses in a voltage indicator can cause unreliability and any problem with the fuse could cause the voltage indicator not to indicate a hazardous voltage. In reality this risk is very small and correct proving procedures will identify the problem. None the less, because of this potential problem, BS EN61243-3 prohibits the use of fuses in 2 pole voltage indicators.
In order to fully comply with BS EN61243-3:2010 Martindale has released an updated version of VI13700/2 with a model number of VI13800. This new model has the fuse replaced by a high wattage resistor in the probe that will limit the current in the event of damage to the cable. This protection is superior to the protection offered by the fuse as the current that can flow under fault conditions is considerably less. The vast majority of competitor products have no protection built into the probe.
HSE Guidance note GS38 has not been revised for many years and therefore hasn’t kept pace with technology or international standards. It is due to be updated. The existing version of GS38 does not require fuses in a 2 pole voltage indicator built to a recognised standard such as BS EN61243-3.
Apart from the fuse/resistor in the probe everything else about the specification of VI13700/2 & VI13800 is identical, including the contrasting inner cable sheath to quickly identify cable damage.
In order to distinguish VI13700/2 from VI13800 the new model has the probe changed to all black instead of yellow. The probe is clearly marked “impedance protected” rather than showing a fuse rating. It is sealed and has no user replaceable parts.
There is no specific requirement to replace the VI13700 products which are in use. NB Proper proving procedures should always be used with any voltage indicator.
- Nominal Voltage Range: 50 - 600V AC / DC
- Nominal Voltage Threshold Indications: 50, 100, 200, 400 V DC/AC rms
- Nominal Voltage threshold tolerance: < 85% of marked threshold indication
- Polarity indication: from approx 12V DC/AC rms
- AC/DC voltage detection: automatic
- Range detection: automatic
- Response time: < 0.1s
- Frequency range: DC, 1 - 400 Hz
- Test current: < 3.5mA at 600V DC/AC rms
- Duty ratio: 30s ON (operated) / 240s OFF (recovery)
- Temperature & Humidity (operating & storage): -10°C to 55°C ≤ 85% R.H.
- Altitude: up to 2000m
- IP rating: IP54
- Power: from circuit under test
- Dimensions: 205(L) x 67(W) x 27(D) mm
- Weight: 130g approx.
- Includes: instructions
- Conforms to BS EN 61243-3:2010 CAT IV 600 V, CAT III 1000V
- Class II, Double Insulation
- Pollution Degree: 2
- EMC: Conforms to BS EN 61326-1